Leave Bluetooth range while connected gauges are logging
The FieldLab Mobile app allows you to remote control logging on Bluetooth-enabled Ralston Gauges, monitor live readings, and harvest test data while within Bluetooth range, which is around 328 feet (100 meters) max.
If you take your mobile device out of range while a gauge is logging, the Field Gauge will continue to log data at the interval you set for as long as it has battery power and until the memory is full.
When you return your mobile device to within Bluetooth range of the gauge and want to reconnect:
- Select the Scan button on the Devices Screen.
- When your gauge appears on the screen, select Connect.
Change Logging Interval
1. Select the Interval Settings Icon in the Logging Interval field. Note: Logging Interval Settings cannot be changed while logging. 2. Select the drop-down menu to choose your new Logging Interval. Note that longer intervals will give you more time ...
Connect to a Bluetooth-enabled Field Gauge with the FieldLab Mobile App
1. Power-on the Field Gauge As soon as you power on your Field Gauge, it will begin advertising its availability to connect to your FieldLab Mobile App. 2. In the FieldLab Mobile App on your mobile device, Select the Scan Button 3. Watch the Devices ...
Get started with FieldLab Mobile
Check compatibility of your mobile device FieldLab Mobile runs on: iOS versions 11.0 and newer. Android versions 9 and newer. Power on your Field Gauge FieldLab Mobile works with any Bluetooth enabled Ralston Field Gauge. Download FieldLab Mobile ...
Get FieldLab Mobile for the iPhone and iPad
Connect to and control your Ralston LC20 Field Gauges with Bluetooth, log and harvest your test data, and share logged data with your organization or client. Download on the App store
A Gauge is Overpressurized
Before applying pressure to your Ralston gauge, confirm that the gauge is rated for higher than the max pressure that will be applied. Gauges that have been overpressured above 33% of max rating will be unable to display or log accurate pressure ...